|Name:||Impulse Chiropractic Adjusting Instrument||Function:||Reduce Pain, Chiropractic Adjusting|
|Usage:||Hospital, Rehabilitation Center, Massage Clinic||Application:||Cervical Syndrome,Lumbar Syndrome|
|Ambient Temperature:||0℃~+45℃||Relative Humidity:||≤85%|
|Power Supply:||AC220V±22V, 50Hz±1Hz||Preloading Force:||5-30N|
400N Impulse Chiropractic Adjusting Instrument,
IQ Impulse Chiropractic Adjusting Instrument,
400N activator adjusting instrument
|Impact||The impact force can be adjusted in five levels. The maximum impact force of each level is: Level I: 100N, Level II: 200N, Level III: 300N, Level IV: 400N, Level V: 500N, tolerance ±50N|
|Stability of impact force||The stability of the impact force generated by the product should be ±20%|
|Output frequency||The output frequency can be adjusted in six levels, the frequency of level I: 2Hz, the frequency of level II: 4Hz, the frequency of level III: 6Hz, the frequency of level IV: 8Hz, the frequency of level V: 10Hz, the frequency of level VI: 14Hz, the tolerance is ±10%|
|Hits||When the automatic mode is not turned on, the number of hits is divided into unlimited times, 5 times, 10 times, 15 times, 20 times, 25 times, 30 times, 35 times, 40 times, 45 times, 50 times, a total of 11 levels adjustable; automatic mode Down, the number of hits each time appears randomly in the range of 10 to 30 times|
|Power supply||AC220V±22V, 50Hz±1Hz|
|Rated input power||300VA|
Contraindications of Impulse Chiropractic Adjusting Instrument
Pictures of Impulse Chiropractic Adjusting Instrument
Accessory loading and unloading
1) Accessory connection
(1) Connect the vibration gun and the host, and select the corresponding vibration head as required;
(2) Connect the electric muscle vibrator to the AC220V, 50Hz power socket through the power cord.
2) Disassembly of accessories
(1) Remove the power cord from the electric muscle vibrator and the power socket;
(2) Remove the vibrating gun from the host and take off the vibrating head.
During transportation, avoid rain and snow splash, vibration and mechanical collision.
The packaged product should be stored in a clean room with a relative humidity of no more than 85%, a temperature of -10°C to 40°C, no corrosive odor, and a well-ventilated, clean room.
Check and clean
|1) Before inspection and cleaning, turn off the power switch and unplug the power cord and wait for 1 minute to fully release the internal capacitor storage to ensure that the product completely stops working.|
|2) Do not immerse any part of the product in water.|
|3) Regularly check the product shell and display screen for cracks or other damage.|
|4) Regularly check whether the accessories are worn or damaged.|
|5) Verify that the connection between the accessory and the host interface is firmly in place.|
|6) Check whether the mainframe enclosure is well sealed.|
|7) When the product is not used for a long time, all accessories should be packed and|
|8) When removing the accessory after use, you should operate it gently and do not pull the accessory cable forcibly, otherwise the accessory cable may be damaged.|
In the short-term final processing of the product and its related accessories, it is necessary to consider the small amount of lead in the solder, and the environmental protection policy of the local government should be followed when processing, and it must not be replaced at will.
Contact Person: Carden