|Name:||Chiropractic Impulse Gun||Application:||Cervical Spondylosis ,lumbar Disc Herniation,|
|Ambient Temperature:||0℃~+45℃||Relative Humidity:||≤85%|
|Atmospheric Pressure:||700hpa~1060hpa||Power Supply:||AC220V 50Hz|
|Rated Input Power:||300VA||Preloading Force:||5-30N|
400N Chiropractic Impulse Gun,
Torticollis Chiropractic Impulse Gun,
400N chiropractic adjusting instrument
Usage of Chiropractic Impulse Gun
1, cervical spondylosis : insomnia, forgetfulness, drowsiness, ringing in the ears, neck tight, wrinkled
or stiff pain, arm numbness, shoulder pain and so on.
2, lumbar disc herniation, spondylolisthesis, spinal stenosis, sciatica, lumbar spine bone spurs, etc.
3, thoracic disease: back pain, chest tightness, stitch, shortness of breath, palpitation and so on.
4, Body Treatment: kyphosis, torticollis, oblique, Bottom, lymphatic massage, beauty chest
Working principle of Chiropractic Impulse Gun
The Chiropractic Impulse Gun generates a time-varying magnetic field by pressurizing the coil with high-voltage pulses, and uses the impact force generated by the core vibration vibration head to treat human lesions.
The electric muscle vibrator sets the corresponding frequency, strength, frequency limit or automatic mode through the host to control the vibration gun, and then the vibration gun performs data processing, and controls the corresponding unit to perform the corresponding output.
Main performance index
|Impact||The impact force can be adjusted in five levels. The maximum impact force of each level is: Level I: 100N, Level II: 200N, Level III: 300N, Level IV: 400N, Level V: 500N, tolerance ±50N|
|Stability of impact force||The stability of the impact force generated by the product should be ±20%|
|Output frequency||The output frequency can be adjusted in six levels, the frequency of level I: 2Hz, the frequency of level II: 4Hz, the frequency of level III: 6Hz, the frequency of level IV: 8Hz, the frequency of level V: 10Hz, the frequency of level VI: 14Hz, the tolerance is ±10%|
|Hits||When the automatic mode is not turned on, the number of hits is divided into unlimited times, 5 times, 10 times, 15 times, 20 times, 25 times, 30 times, 35 times, 40 times, 45 times, 50 times, a total of 11 levels adjustable; automatic mode Down, the number of hits each time appears randomly in the range of 10 to 30 times|
|Noise||When the product is working normally, the noise is not more than 80dB (A)|
|Power supply||AC220V±22V, 50Hz±1Hz|
|Rated input power||300VA|
Pictures of Chiropractic Impulse Gun
Accessory loading and unloading
1) Accessory connection
(1) Connect the vibration gun and the host, and select the corresponding vibration head as required;
(2) Connect the electric muscle vibrator to the AC220V, 50Hz power socket through the power cord.
2) Disassembly of accessories
(1) Remove the power cord from the electric muscle vibrator and the power socket;
(2) Remove the vibrating gun from the host and take off the vibrating head.
During transportation, avoid rain and snow splash, vibration and mechanical collision.
The packaged product should be stored in a clean room with a relative humidity of no more than 85%, a temperature of -10°C to 40°C, no corrosive odor, and a well-ventilated, clean room.
Contact Person: Amanda